Output power set – increments and decrements in 5W steps.
Reflection power trip set – if a process requires the microwave power to be disabled when a certain reflected power level is detected, this function will allow that trip value to be set. This function increments and decrements in 1W steps.
Timer set – the system has the option of being used in continuous or [timer] mode. When the microwave power is enabled, the timer function will decrement in 1 second steps and disable the microwave power when the timer reaches zero.
[START/STOP] – enables and disables microwave power.
[SWEPT] – when used in standalone mode, this mode allows an improved accuracy measurement of real-time reflected power to be made and displayed.
[TIMER] – when the user requires to have power automatically disabled after a predetermined test duration.
[EXTERNAL SOURCE] – for use with users own signal generator via SMA female connector on the rear panel.
[USB] – When USB communication is achieved, this button will automatically illuminate.
[RESET] – resets the system back to the power-on values.
[DATA CAPTURE] – If data is required for analysis, this feature allows all the system parameters including settings, status and measurement to be easily captured.
With [START/STOP] enabled, the system displays the real-time measurement of the output and reflected power as well as the amplifier temperature.